The application of leveling bubble to the portable laser beam instrument
Place the three foot frame in about 1.5 meters before the test. The lower extremities can be measured with a height of one section, and three nodes of the spine can be selected. When the ground is not flat, the length of the foot of the three foot should be adjusted so that the blister in the bubble meter is in the red ring (figure I) the foot is in a horizontal state.
Ks_ laser alignment instrument is a high and new technology product developed by many years on the basis of foreign advanced technology. It is effective and reasonable to use the principle of laser linearity and accurately measure 0 degree - 360 degree arbitrary angle. The product is a product designed for professional prosthetic assembly technicians and lower limbs and spinal orthopedics. It is beautiful in shape, clear in line, with high precision and easy to operate, and is convenient to carry. It provides a scientific and accurate measuring line for rehabilitative orthopedics and artificial limb assembly. It is a necessary precision measuring instrument for artificial limb orthopedics.
1. the machine uses a vertical line laser tube and a rotating line laser tube, and the laser tube is wavelength 635nm.
The 2. vertical laser line is adjusted by the level adjustment of the cloud platform or the small knob adjustment, and the rotating laser line is adjusted up and down from the handle of the cloud platform.
3. the upper end of the fuselage is equipped with an illuminated scale plate and a vertical laser line level.
The 4. laser alignment instrument is equipped with a special charger and a charging socket. When the charging is on the machine, the yellow light is bright, the red light is on the charge and the green light is bright when the battery is saturated. (Note: the charging is strictly prohibited to prevent damage to the laser tube.)
5. the machine assembly three section 5 1.2 volt 1300 Ma rechargeable battery, one charge can work for 15 hours in a row.
6. instruments wear special level three foot frame and three axis adjustable cloud platform. It can be adjusted arbitrarily.
2. special charger
3. three foot frame
4. special aluminum alloy suitcase
5. three section 5 1.2 volt 1300 Ma rechargeable battery
1. to place the three foot frame in about the front of the subject. The lower extremities can be measured with a height of one section, and three nodes of the spine can be selected. When the ground is not flat, the length of the foot of the three foot should be adjusted so that the bubble in the bubble meter is in the red ring (Figure 1) in a horizontal state.
2. the fixed plate and the seat of the cloud platform are connected and fixed (Figure two), and then installed on the platform of the cloud platform, the locking button is pressed and fixed (Figure three).
3., adjust the level of the Yuntai, the rotation shaft, the tightening screws, the upper and lower rotary axes to tighten the handle, (adjust the proper size), adjust the fine adjustment knob, and place the level bubble on the line instrument in the middle (Figure four) and fasten the knob 4.
4. to start.
Two laser lines and dial display, respectively the adjusting knob 6 vertical laser line and the laser line re synthesis of horizontal rotating a vertical line, two line intersection is the center of rotation instrument.
5. replacement of battery method
Pull the rear lid knob with laser, Cross sent to open the back cover to replace the battery.